top layer filtering sampler
KomrisFiltr Ltd., Zelenograd,
tel./fax +7 499 710 57 27,
E-mail : firstname.lastname@example.org<![if !supportNestedAnchors]><![endif]> , email@example.com
www.1filtr.ru , http://www.mtu-net.ru/comrisfilter<![if !supportNestedAnchors]><![endif]>
TOP LAYER FILTERING SAMPLER OFF-25.
The device is intended for preparing the sample on the transparent track membrane
for microscopy in flotation investigation, for example, in parasitology.
Structure and Work
scheme of device is shown in the drawing. The apparatus is placed on the rack 1,
equipped with the bean (is not shown in the drawing) for connection to suction,
vacuum-gauge 2, rarefaction regulator 3, ball tap 4 with the female conical
connector 5 for the male conic 6 of the base 7. The membrane 8 (disk
The device works by the next way. Rarefaction is produced in the space under membrane 8 by a suction device (waterjet pump, surgical aspirator pump, vacuum-compressor etc.). When the regulator 17 is open, air passes in the cylinder 9 through pipe 18 and further from the cylinder in the space under membrane 8. If the pipe 18 end is immersed in a liquid, last is sucked into the cylinder 9 over the membrane 8. If the regulator 17 to be closed, the liquid over membrane 8 is filtered through it, leaving off microparticles on it. Left on the membrane 8 microparticles may be processed (rinsed, fixed, dyed, dried etc.) according a used method by the same feeding processing solutions onto the membrane 8. After the processing the membrane is placed on the slide for microscopic investigation.